Each Complete Kit is supplied with a Reader unit. One Reader can be used to control as many doseBadges as you need.
The Reader is used to carry out the following functions:
Configuring the doseBadge
The Reader unit can also be used to configure the doseBadge settings. The doseBadge can be configured to meet the ISO Q=3 (for European regulations) and OSHA Q=5 (for the American standards).
The Exchange Rate, Criterion Level, Criterion Time, Threshold and Slow Time Weighting can be set individually in order to meet other standards such as those published by MSHA, AICHE and ACGIH. See the Noise Regulations page for more information.
As the doseBadge has Dual Channels, it can have Channel 1 set to OSHA, MSHA, AICHE, ACGIH or similar, with Channel 2 fixed to measure with the common ISO 3dB settings.